300 Piece/Pieces per Month Repair Alma Soprano 810nm handpiece tools kit
Packaging & Delivery
Plastic packaging case for Repair Alma Soprano 810nm handpiece tools kit with Germany Jenoptik laser bar
Repair Alma Soprano 808nm 810nm handpiece tools kit with Germany Jenoptik laser bar
Alma laser hair removal handpiece repair
We have more than 16 years mounting experience, our engineer assembled the diode laser chips in our own clean lab, Our laser bars are imported from Germany with high quality and good stability
Diode laser handle piece and laser stack repair process:
First, the customers need to send us the old diode as below pictures shows, if can’t send old diode laser, need send to us structure design of the diode and dimension details stack. We will repair the old diode or design a new stack to replace the old one.
Second, we disassemble the handle piece, and test the old laser stack. For the micro channel laser stack, Such as Alam laser, Lightsheer, Syneron , we will test each laser plate one by one to confirm whether there are any one bars still can work. If there have some good bars, the repairing charge will lower. For macro laser stack like most product made in China, we will test whole laser stack together.
Third, we choose best quality laser stack to replace the old broken one. Our bars are imported from Germany. We mounted bars and assemble them to laser stack in our lab.Every stack will be fully tested and had 8 hours buring test.
Finally, our experienced technician will reassemble the handle piece. Which include:
a: Install the new laser stack into the handle piece
b: Clean lens and check the TEC and other parts inside the handle piece
c: Test the power output of whole handle piece
d: Clean the cover shell of handle piece and package it.
We also professional other diode laser stack for laser handpiece.